Cleaning condenser mic capsule
What liquid best to use on a (true) condenser mic capsule to clean -
grime and dust, all well-stuck on ? Iso, distilled water, borax !?
And for that matter, same question for electret ?
Is the gold evaporation on the outside or inside, so what level of
delicacy - extreme, or extremely extreme ?
Capsule is CK12 - no, not a brass-ring one, sadly. Also some CK1 etc.
Cleaning condenser mic capsule
On 9/11/2020 7:28 AM, geoff wrote:
> What liquid best to use on a (true) condenser mic capsule to clean -
> grime and dust, all well-stuck on ? Iso, distilled water, borax !?
Distilled water or alcohol, on a soft brush. It's been suggested that
steam be used to deposit a proper amount of water on the brush. DO NOT
USE STEAM TO CLEAN THE DIAPHRAGM!
> And for that matter, same question for electret ?
> Is the gold evaporation on the outside or inside, so what level of
> delicacy - extreme, or extremely extreme ?
It's on the outer surface of the diaphragm, so be extremely cautious
> Capsule is CK12Â* - no, not a brass-ring one, sadly. Also some CK1 etc.
Best advice is not to try it on a high quality microphone. Better
practice on something cheap first. Best is to send the mic back to the
manufacturer or a trusted repair person who has enough experience to not
screw it up.
David Josephson (Josephson Microphones) used to have an article on his
web site about cleaning a mic diaphragm that he distilled from an
on-line discussion in rec.audio.pro. Maybe some of that is still in an
archive if you want to try to find it.
David decided that too many people were ruining too-good microphones by
attempting to clean the diaphragm and ultimately removed the article
from his web page, replacing it with this:
> More than twenty years ago, we posted some comments to rec.audio.pro about
> cleaning condenser microphone diaphragms. I have decided to retire the local
> copy of that article, because it seems to inspire too many to attempt things
> that are unwise.
> One thing left out of the original article is this: cleaning should only be
> done if the microphone fails to work otherwise. The difference caused by a
> little extra weight may be audible, but not huge. Got that? No cleaning unless
> it *doesn't work.* And in no case should you use any source of heat near the
> microphone. Someone thought that what we wrote suggested putting steam on the
> microphone diaphragm. !!?! ... the steam is to put a slight amount of pure
> water on your cleaning brush, not the diaphragm! No heat anywhere near
> diaphragm, OK?
> With all due respect to others in the field, if the conductive layer on
> the diaphragm is so fragile that gentle cleaning with distilled water, alcohol
> and cotton or a slightly damp fine brush damages it, you have no business
> cleaning it. That said, losing a small amount of metallization has little or
> no effect -- the diaphragm will still behave as it did before, the
> active area is just reduced a little. A real hazard, however, is that the
> capsule will become contaminated internally with whatever the cleaning medium
> carries with it, and will forever be noisy. This is particularly a problem with
> older PVC diaphragms that have cracks. Unless you have a way to test for
> ionic contamination of the surface and resulting leakage resistance of the
> capsule, don't try it.
> If any Josephson microphone appears to need cleaning, please send it back to
> the factory in California. The cost is minimal and you can be sure of original
> performance when it's done.
For a good time, call http://mikeriversaudio.wordpress.com
|Thread||Thread Starter||Forum||Replies||Last Post|
|Help With Condenser Mic Capsule ID||Mark Robinson||Pro Audio||3||October 5th 05 04:09 PM|
|??? Electret Condenser Capsule???||Serge Stodolnik||Pro Audio||3||February 14th 05 02:35 PM|
|Large capsule condenser||Vland||Pro Audio||3||January 21st 05 08:20 AM|
|(MIC) --AKG c460 b with ck5 condenser capsule--whats a fair price??||Joseph||Pro Audio||0||May 18th 04 01:58 AM|
|condenser mic capsule impedance and tube circuitry||xy||Pro Audio||21||October 8th 03 07:20 AM|